Model No.: cx
Measurement is not affected by the variation of flow density, viscosity, temperature,
pressure and conductivity. High accuracy measurement is guaranteed according to the
linear measurement principle.
No obstacle in the pipe, no pressure-loss and lower requirement for straight pipeline.
DN 6 to DN2000 covers a wide range of pipe size. A variety of liners and electrodes are
available to satisfy different flow characteristic.
Programmable low frequency square wave field excitation, improving measurement
stability and reducing power consumption.
Implementing 16 bits MCU, providing high integration and accuracy; Full-digital
processing, high noise resistance and reliable measurement; Flow measurement range up
High definition LCD display with backlight.
RS485 or RS232 interface supports digital communication.
Intelligent empty pipe detection and electrodes resistance measurement diagnosing
empty pipe and electrodes contamination accurately.
SMD component and surface mount technology (SMT) are implemented to improve the
WELTER electromagnetic flowmeter can be used to measure the volume flow of
conductive fluid in a closed pipeline. It is widely applied in the flow measurement and
control in the fields of chemical and petroleum industry, metallurgy industry, water and
waste water, agriculture and irrigation, paper making, food and beverage industry and
Ambient temperature: sensor: -25C to + 60C; converter: -25C to + 60C.
Relative humidity: 5% to 90%;
Maximum fluid temperature:
Compact type: 60C
Remote type: Teflon 150C
Neoprene 80C; 120C
Fluid conductivity: 5μS/cm
2. WORKING PRINCIPLES
2.1 Measuring Principles
The measuring principle of electromagnetic flowmeter is based on the electromagnetic
induction law of Farady. The sensor is mainly composed of measuring tube with isolate lining, a
pair of electrodes installed by penetration of the measuring tube wall, a pair of coils and iron core
to produce working magnetic field. When the conductive fluid flows through the measuring tube
of the sensor, the voltage signal in direct proportion to the average flow velocity of the fluid will
be inducted on the electrodes. The signal is amplified and treated by the transmitter to realize
various display functions.
2.2 Converter Circuit Schematic
The converters supplies a stable exciting current to the coil in the sensor of electronetic
flowmeters to get B constant and amplifies the electromotive force and convert it into standard
signals of current or frequency so that the signals can be used for displaying, controlling and
3. TECHNICAL SPECIFICATION
3.1The flowmeter is compliant to Standard "JB/T 9248-1999 Electromagnetic Flow
3.2Maximum flow speed: 15m/s
3.3DN size: 3, 6, 10, 15, 20, 25, 32, 40, 50, 65, 80, 100, 150, 200, 250, 300, 350, 400, 450,
500, 600, 700, 800, 900, 1000, 1200, 1400, 1600, 1800, 2000
3.4Accuracy: 0.5%, 0.2%
3.5Nominal Pressure: 4.0MPa(DN0-150)
Or other specified by order
Electrode Forms and Materials:
Electrode design has four forms: standard form, scraper form, detachable form andgrounding electrode.
Electrode material is selectable from stainless steel containing Mo, stainless steel coated with
carbonized tungsten, Hastelloy B, Hastelloy C, Titanium, Tantalum and platinum-iridium alloy.
Flange material: carbon steel
Grounding ring: stainless steel
Inlet protection ring: carbon steel, stainless steel
IP65;IP68: only available for remote-type sensor with neoprene or polyurethane liner, not
including flameproof model. Contact us if you need more details on flow meter. We are ready to answer your questions on packaging, logistics, certification or any other aspects about water flow meter、magnetic flow meter. If these products fail to match your need, please contact us and we would like to provide relevant information.
Premium Related Products