Model No.: cx
Name and unit
|Name and unit||Model|
|Top flow (m3)||100||100|
|Min. scale value(m3)||0.8×10-5||0.2×10-4|
|Normal pressure (Pa)||1000||1000|
|Accuracy in rated flow (%)||±1||±1|
|Pressure in rated flow( Pa≤)||80||80|
|Outline dimensions(L/W/H)||490×264×220 mm||490×264×304 mm|
XML type is normal-type, it adopts brass material, and generally used in the scope of non-corrosive gases.
XMF type is anticorrosive-type, it adopts stainless steel material, can measure corrosive gases.
1. Put the meter on the worktable, adjust screws to enable blister of leveler in the centre, and maintain for a long time in use.
2. Open sealed screw cap of water level controller, pull out internal wool rope.
3. In the plug jack of thermometer or manometer, inject distilled water into the meter, when redundant distilled water drain from hole of water level controller ( sometimes it is clean when outflow one drop), then put wool rope in sealed screw cap of water level controller, and fasten the sealed screw cap.
4. Install thermometer and manometer, (every case 10Pa)
5. Connect air channel according to direction of gas outlet and intake, and sealled.
6. Open gas valve, and then you can measure gas.
IV. Attentions while using
1. When you use it, you must always pay attention to the water level in the meter, otherwise it will affect measuring accuracy.
2. When you use it, temperature should be maintained at 15 ~ 25 ºC, and the temperature is the same as room temperature, its temperature difference should be ≤ 2 ºC.
3. You'd better readout in a few weeks after the operation of the meter.
4. When pressure of measuring gas is go beyond normal pressure 1 ~ 6 times (1 ~ 6 KPa), the meters can still work, at this time manometer should be removed , then use rubber plug plugs up mounting holes of manometer.
5. When flow of measuring gas is go beyond rated flow, the meters can still work, but measuring accuracy will be greatly decreased.
7. The meter should not be installed in a cold room, for fear that it will freeze inside.
8. In normal circumstances, check out accuracy every year at least.
1 pc of thermometer
1 pc of manometer
2 rubber plugs
Payment Terms of electromagnetic flow meter
1.Loading Port: Shanghai China
2. Samples Leading Time: 3-7 working days
3. Delivery Time: Within 7-15 days after payment
4.Payment: T/T, Western Union, Paypal
After-sales service of electromagnetic flow meter
We promise to deliver the following after-sales service:
1. Warranty: The warranty period is 12 months starting from the shipping date;
2. Customers will enjoy free repair services for failures found within the warranty due to manufacturing quality;
3. For on-site services not arising from manufacturing quality, we shall charge a certain service fee at our discretion based on actual charges incurred such as travel expense;
4. We provide paid repair services for meter damages not arising from manufacturing quality issues or meter damages not within the scope of warranty. Contact us if you need more details on Gas Flow Meter. We are ready to answer your questions on packaging, logistics, certification or any other aspects about Natural Gas Flow Meter、Nitrogen Gas Flow Meter. If these products fail to match your need, please contact us and we would like to provide relevant information.
Premium Related Products